Joint continuous passive motion instrument (elbow vertical)
Joint continuous passive motion instrument: 01. When the rehabilitation device was started, the stretching movement appeared first, and the angle (α) between the fixed bracket and the movable bracket moved to 180 degrees (the initial position of movement was when α = 180 degrees). 02. The adjustable range of the length of the mobile support of the rehabilitation device shall not be less than 50 mm. 03. The range of angle (α) movement between the fixed support and the movable support of the rehabilitation device should not be less than 100 °
04. The rated load of the rehabilitation device is 50N, and it should be able to work smoothly without jamming under the rated load, and there is no abnormal impact sound during reciprocating operation.
05. The working noise of the whole machine should not be greater than 65dB. 06. The adjustable range of angle (α) between the fixed and movable support of the rehabilitation device: the low speed is not more than 0.5 ° / s, and the high speed is not less than 3.5 ° / s, which can be adjusted in nine gears. 07. The rehabilitation device can work continuously for more than 4H under 50N load. 08. A manual controller is set up for the rehabilitation device, so that the patient can control the recovery device to pause or reset (α to 180 ° movement). 09. The input power of the rehabilitation device is 70va. 10. Rated working voltage: 220 VAC ± 22 VAC